Durability Test of a Developed Radon Counter Module

Radon is a natural, inert, invisible, odorless and chemically inactive radioactive gas emitted from the earth. Because inhaling radon and its radioactive decay products causes irradiation of lung tissue, prolonged exposure to a high concentration of radon significantly increases the risk of developing cancer. For our experiments, we implemented a radon with a PIN photodiode radon-sensor module. Th radon counter was equipped with a Wiwireless communication and radon monitoring. In order to improve the performance of the developed radon counter module, the durability test was performed.

invisible, odorless and emitted from the Because inhaling radon and its radioactivedecay products causes irradiation of lung tissue, prolonged exposure to a high concentration of radon significantly increases the risk of developing cancer.
implemented a radon counter sensor module. The -Fi module for wireless communication and radon monitoring. In order to improve the performance of the developed radon counter module, the durability test was radon, high concentration, PIN Radon is a chemical element with symbol Rn and 86. It is a radioactive, colorless, odorless, tasteless noble gas. It occurs naturally as an intermediate step in the normal radioactive decay chains through which thorium and uranium slowly y into lead; radon, itself, is a decay product of Rn, has a half-life of 3.8 days. Since thorium and uranium are two of the most common radioactive elements on Earth, and since their isotopes have very long half-lives, on the order of billions of years, radon will be present long Unlike all the other intermediate elements in the decay chains, radon is, under normal conditions, gaseous and easily inhaled. Radon gas is a There are many commercial instruments and techniques available for measuring radon indoors. Most detectors for evaluating indoor radon levels are passive in that they do not require external power. The principal drawback of passive detectors is that they only measure radon concentration at one specific location for a specific period. Because many variables influence radon concentration levels, a single estimate of radon concentration is likely to have a significant error. In [1], highly sensitive, electrostatic collection chambers using Columbia Resin 39 (CR track detectors were developed for measuring low level radon. In [2], the unique characteristics and the shortcomings of track detectors for neutron and radon dosimetry is described and compared with those of alternative detector systems. In [3], low particle sensor systems using special Oxide Semiconductor (PMOS) transistors in a floating n-well were developed for radon and radon monitoring and dosimetry. In [4], a radon detect employed an electrically charged, pressed, porous metal filter that permitted radon blocking ambient light. It readily trapped both attached and unattached Polonium Polonium 218 (Po-2l8) ions present in gas passing through the filter. The filter was positively charged relative to the unbiased PN junction of a photo detector within a detection chamber. In [5], an alpha particle detector was designed using a commercial silicon photodiode. An application of a silic photodiode detector for radon progeny measurements is presented in [6]. Most detectors for evaluating indoor radon levels are passive in that they do not require external power. The principal drawback of passive detectors is that they ncentration at one specific location for a specific period. Because many variables influence radon concentration levels, a single estimate of radon concentration is likely to have a significant error. In [1], highly sensitive, electrostatic collection bers using Columbia Resin 39 (CR-39) plastic were developed for measuring lowlevel radon. In [2], the unique characteristics and the shortcomings of track detectors for neutron and radon dosimetry is described and compared with those of ternative detector systems. In [3], low-cost alphaparticle sensor systems using special Positive Metalemiconductor (PMOS) transistors in a floating well were developed for radon and radon-daughter monitoring and dosimetry. In [4], a radon detector employed an electrically charged, pressed, porous metal filter that permitted radon-gas diffusion while blocking ambient light. It readily trapped both attached and unattached Polonium-214 (Po-214) and 2l8) ions present in gas passing hrough the filter. The filter was positively charged relative to the unbiased PN junction of a photo-diode detector within a detection chamber. In [5], an alpha particle detector was designed using a commercial silicon photodiode. An application of a silicon photodiode detector for radon progeny measurements International Journal of Trend in Scientific @ IJTSRD | Available Online @ www.ijtsrd.com In this paper, we describe a low-end real time detector that uses a PIN photodiode radon a Wi-Fi wireless communication performance of the developed radon counter module could be improved through a variety of durability tests.

II. Pin Photodiode Radon Counter
A commercial PIN photodiode can be used to detect radiation, and particularly, alpha particles cost, has good quantum efficiency, and go resolution. It can also work with a low bias voltage. A PIN photodiode is more widely used than a conventional photomultiplier tube (PMT) because it requires less biasing to operate and it is very compact. The Mega 2560 is a microcontroller board the Atmega 2560. It has 54 digital input/output pins (of which 15 can be used as Pulse Width (PWM) outputs), 16 analog inputs, 4 UARTs (hardware serial ports), a 16 MHz crystal oscillator, a USB port, a power jack, an inprogramming (ICSP) header, and a reset button. It contains everything needed to support the microcontroller; it just needs to be connected to a computer with a USB cable or powered with an AC to-DC adapter or battery. The Mega 2560 board is compatible with most shields designed for the Uno and the older Duemilanove and Diecimila boards.
In our experiments, we used a radon counter assembled from very low-cost consumer electronics. The circuit designs for the power, high voltage generator, LCD, switch, LED, buzzer, microprocessor control unit (MCU), and sensor circuit Using these circuit elements, we developed the PIN photodiode radon-counter printed circuit board (PCB) layout and then assembled the PIN photodiode radon counter shown in Fig. 1.
The ESP8266 Wi-Fi module is a self system-on-chip (SOC) with integrated TCP/IP protocol stacks that can give any microcontroller access to a Wi-Fi network. The ESP8266 is capable of either hosting an application or offloading all Wi networking functions to another application processor. Each ESP8266 module comes pre-programmed with AT Command Set firmware, meaning the module can be hooked up to an Arduino device with about the same Wi-Fi ability as a Wi-Fi shield. The ESP8266 module is an extremely cost-effective board with a huge, and ever growing, community of users. This International Journal of Trend in Scientific Research and Development (IJTSRD) ISSN: 2456 @ IJTSRD | Available Online @ www.ijtsrd.com | Volume -2 | Issue -2 | Jan-Feb end real time radon detector that uses a PIN photodiode radon-sensor with Fi wireless communication module. The radon counter module could be improved through a variety of durability A commercial PIN photodiode can be used to detect particles. It is low cost, has good quantum efficiency, and good energy resolution. It can also work with a low bias voltage. A PIN photodiode is more widely used than a conventional photomultiplier tube (PMT) because it requires less biasing to operate and it is very compact.
he Mega 2560 is a microcontroller board based on It has 54 digital input/output pins idth Modulation (PWM) outputs), 16 analog inputs, 4 UARTs (hardware serial ports), a 16 MHz crystal oscillator, a -circuit serial programming (ICSP) header, and a reset button. It contains everything needed to support the microcontroller; it just needs to be connected to a computer with a USB cable or powered with an AC-DC adapter or battery. The Mega 2560 board is h most shields designed for the Uno and the older Duemilanove and Diecimila boards.
In our experiments, we used a radon counter cost consumer electronics. The circuit designs for the power, high voltage , buzzer, microprocessor control unit (MCU), and sensor circuit were made.
, we developed the PIN counter printed circuit board (PCB) then assembled the PIN photodiode radon Fi module is a self-contained chip (SOC) with integrated TCP/IP protocol stacks that can give any microcontroller Fi network. The ESP8266 is capable of either hosting an application or offloading all Wi-Fi g functions to another application processor. programmed with AT Command Set firmware, meaning the module can be hooked up to an Arduino device with about the Fi shield. The ESP8266 effective board with a huge, and ever growing, community of users. This module has powerful enough on storage capability to allow it to be integrated with the sensors and other applicationits general-purpose input/outputs (GPIOs), with minimal development up-front and minimal loading during runtime. Its high degree of on allows for minimal external circuitry. The photodiode radon counter with t module is shown in Fig. 2.  Page: 109 module has powerful enough on-board processing and storage capability to allow it to be integrated with the -specific devices through purpose input/outputs (GPIOs), with front and minimal loading during runtime. Its high degree of on-chip integration allows for minimal external circuitry. The PIN photodiode radon counter with the ESP8266 Wi-Fi Implemented PIN photodiode radon counter  The temperature cycle test and the thermal shock test were performed to evaluate the ability of the radon counter to resist extremely low and extremely high temperatures, as well as the ability of the radon counter to withstand cyclical exposures to these temperature extremes. For temperature cycle test, the radon counter powered with the adapter was evaluated at temperature of -20 °C for 1 hour, and then evaluated at temperature of 70 °C for 1 hour, continuously. This process was repeated total 5 times.
For the thermal shock test, the radon counter which was not powered was evaluated at temperature of -40 °C for 30 minutes, and then evaluated at temperature of 80 °C for 30 minutes, continuously. This process was repeated total 50 times. The vibration withstand test was performed by evaluating the radon counter which was not powered was evaluated at two vibration frequencies with an oscillation amplitude of 1.5 mm. Vibration frequencies were 10 Hz and 50 Hz, which were changed minutely. This process was performed for 2 hours in the X, Y, and Z axis directions, respectively. HALT is a test technique called test-to-fail, where a product is tested until failure. HALT was performed to evaluate operating limits of the radon counter for lower and upper temperatures, and vibration. If any alteration was not detected at an initial temperatures and vibration for 5 minutes, the test was repeated at increased test conditions until the alteration was detected. In the environmental tests, which were the heat resistance, the cold resistance, the humidity resistance, the temperature cycle, the thermal shock, and the vibration withstand test, the evaluation criteria was that there was no abnormality in the appearance, the  LCD display, and the operation of the radon counter. The evaluation criteria for HALT were that operating limits of the radon counter for lower and upper temperatures and vibration were measured and evaluated.
@ IJTSRD | Available Online @ www.ijtsrd.com | Volume -2 | Issue -2 | Jan-Feb 2018 Page: 112 Table 2 shows the testing environment and result for the durability test. For the heat resistance test, we used a high-temperature chamber (Espec, Tokyo, Japan), which could change the temperature from room temperature to 200 °C. A thermo-hygrostat chamber (Espec), which could change the temperature from -40 °C to 200 °C and the relative humidity from 20 % to 98 %, was used for the cold resistance, the humidity resistance, and the temperature cycle test. For the thermal shock test, a thermal shock chamber (Espec), which could change the temperature from -77 °C to 205 °C, was used. For the vibration withstand test and the HALT, we used a HALT chamber (Qualmark, Denver, USA), which could change the temperature from -100 °C to 200 °C within 2 minutes and the vibration from 1 G to 50 G.
Any abnormalities were not observed on the appearance, the LCD display, and the operation of the radon counter in the environmental tests, which were the heat resistance, the cold resistance, the humidity resistance, the temperature cycle, the thermal shock, and the vibration withstand test (Table 2). In the HALT, we observed operating limits of the radon counter, which were the temperatures from -37 ℃ to 72 ℃ and the vibration of 23 G. The LCD display was not operated at the temperature of -37 ℃, and abnormality of LCD display screen was observed at the temperature of 72 ℃. The high voltage chamber of the radon counter was disassembled at the vibration of 23 G.

III. Concluding Remarks
Radon is an invisible, odorless and chemically inactive radioactive gas that is produced by the decay of uranium ore, such as radium, actinium, or thorium. Because inhaling radon and its radioactive decay products causes irradiation of lung tissue, prolonged exposure to high concentrations of radon significantly increases the risk of developing cancer. It has been reported that the US. Environmental Protection Agency estimates exposure to naturally occurring radon leads to 21,000 lung cancer deaths nationwide each year, making radon the nation's primary environmental health threat and second only to cigarette smoking as a cause of fatal lung cancer. Various types of equipment and components have been proposed to date for radon detection. For our experiments, we implemented a radon counter with a PIN photodiode radon-sensor module. The radon counter was equipped with a Wi-Fi module for wireless communication and radon monitoring. We confirmed that the performance of the developed radon counter module was improved through some durability test.